<?xml version='1.0' standalone='yes' ?>
<GXML_Root>
  <Array dim='[2]' type='Cluster'>
    <Signal_conditioner mems='4'>
      <Signal_conditioner_up mems='6'>
        <Name type='String'>M68D1</Name>
        <Manufacturer type='String'>M68D1</Manufacturer>
        <Model_name type='String'>M68D1</Model_name>
        <Serial_number type='String'>25</Serial_number>
        <Inventory_item_number type='String'></Inventory_item_number>
        <Type type='String'>Charge amplifier</Type>
      </Signal_conditioner_up>
      <Transfer_function_Signal_Conditioner mems='3'>
        <Setting type='String'></Setting>
        <Comment type='String'></Comment>
        <Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_ dim='[0,0]' type='DBL'>
          <Numeric_Control></Numeric_Control>
        </Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_>
      </Transfer_function_Signal_Conditioner>
      <Transfer_function dim='[1]' type='Cluster'>
        <Transfer_function_2 mems='3'>
          <Setting type='String'>10mV/pC 0.1 - 20000</Setting>
          <Comment type='String'></Comment>
          <Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_ dim='[4,5]' type='DBL'>
            <Numeric_Control>0.1</Numeric_Control>
            <Numeric_Control>0.01</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>0.0001</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>5</Numeric_Control>
            <Numeric_Control>0.01</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>0.0001</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>10000</Numeric_Control>
            <Numeric_Control>0.01</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>0.0001</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>20000</Numeric_Control>
            <Numeric_Control>0.01</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
            <Numeric_Control>0.0001</Numeric_Control>
            <Numeric_Control>0</Numeric_Control>
          </Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_>
        </Transfer_function_2>
      </Transfer_function>
      <Signal_conditioner_bottom mems='6'>
        <Comment type='String'></Comment>
        <Date_of_previous_calibration type='DBL'>0</Date_of_previous_calibration>
        <Date_of_next_calibration type='DBL'>0</Date_of_next_calibration>
        <Calibrated_by type='String'></Calibrated_by>
        <Minimum_frequency_Hz type='DBL'>0.1</Minimum_frequency_Hz>
        <Maximum_frequency_Hz type='DBL'>20000</Maximum_frequency_Hz>
      </Signal_conditioner_bottom>
    </Signal_conditioner>
    <Signal_conditioner mems='4'>
      <Signal_conditioner_up mems='6'>
        <Name type='String'>BK 2647A</Name>
        <Manufacturer type='String'>Bruell Kjaer</Manufacturer>
        <Model_name type='String'>2647A</Model_name>
        <Serial_number type='String'>2613590</Serial_number>
        <Inventory_item_number type='String'></Inventory_item_number>
        <Type type='String'>Charge amplifier</Type>
      </Signal_conditioner_up>
      <Transfer_function_Signal_Conditioner mems='3'>
        <Setting type='String'></Setting>
        <Comment type='String'></Comment>
        <Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_ dim='[0,0]' type='DBL'>
          <Numeric_Control></Numeric_Control>
        </Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_>
      </Transfer_function_Signal_Conditioner>
      <Transfer_function dim='[1]' type='Cluster'>
        <Transfer_function_2 mems='3'>
          <Setting type='String'>1.001 mV/pC 0.1 - 20000 Hz</Setting>
          <Comment type='String'>Signal conditioner</Comment>
          <Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_ dim='[25,5]' type='DBL'>
            <Numeric_Control>0.1</Numeric_Control>
            <Numeric_Control>0.000774</Numeric_Control>
            <Numeric_Control>-138</Numeric_Control>
            <Numeric_Control>4E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>5</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>6.3</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.2</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>8</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.4</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>10</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.5</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>12.5</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.6</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>16</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.7</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>20</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.7</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>25</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.8</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>31.5</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.8</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>40</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.9</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>50</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.9</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>63</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.9</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>80</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-179.9</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>100</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>800</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.1</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>2000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.2</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>3150</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.3</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>0.5</Numeric_Control>
            <Numeric_Control>5000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.4</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>1</Numeric_Control>
            <Numeric_Control>6300</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.5</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>1</Numeric_Control>
            <Numeric_Control>8000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.7</Numeric_Control>
            <Numeric_Control>5E-6</Numeric_Control>
            <Numeric_Control>1</Numeric_Control>
            <Numeric_Control>10000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-180.9</Numeric_Control>
            <Numeric_Control>7E-6</Numeric_Control>
            <Numeric_Control>2</Numeric_Control>
            <Numeric_Control>12500</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-181.1</Numeric_Control>
            <Numeric_Control>7E-6</Numeric_Control>
            <Numeric_Control>2</Numeric_Control>
            <Numeric_Control>16000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-181.4</Numeric_Control>
            <Numeric_Control>7E-6</Numeric_Control>
            <Numeric_Control>2</Numeric_Control>
            <Numeric_Control>20000</Numeric_Control>
            <Numeric_Control>0.001001</Numeric_Control>
            <Numeric_Control>-181.7</Numeric_Control>
            <Numeric_Control>7E-6</Numeric_Control>
            <Numeric_Control>2</Numeric_Control>
          </Frequency_Hz_-----------Sensitivity_V_per_pC_-----Phase_degrees_>
        </Transfer_function_2>
      </Transfer_function>
      <Signal_conditioner_bottom mems='6'>
        <Comment type='String'></Comment>
        <Date_of_previous_calibration type='DBL'>3.629146E+9</Date_of_previous_calibration>
        <Date_of_next_calibration type='DBL'>3.660682E+9</Date_of_next_calibration>
        <Calibrated_by type='String'>VUT</Calibrated_by>
        <Minimum_frequency_Hz type='DBL'>0.1</Minimum_frequency_Hz>
        <Maximum_frequency_Hz type='DBL'>20000</Maximum_frequency_Hz>
      </Signal_conditioner_bottom>
    </Signal_conditioner>
  </Array>
</GXML_Root>