<?xml version='1.0' standalone='yes' ?>
<GXML_Root>
  <Array dim='[1]' type='Cluster'>
    <Sensor mems='6'>
      <Sensor_cluster_1 mems='13'>
        <Name type='String'>PCB J353B01</Name>
        <Manufacturer type='String'>PCB</Manufacturer>
        <Model_name type='String'>J353B01</Model_name>
        <Serial_number type='String'>145292</Serial_number>
        <Status type='String'>internal</Status>
        <Inventory_number type='String'></Inventory_number>
        <Test_equipment_number type='String'></Test_equipment_number>
        <Type type='String'>ICP</Type>
        <Measurand type='String'>acceleration</Measurand>
        <Sensitivity type='DBL'>0.002014</Sensitivity>
        <Typical_sensitivity type='DBL'>0</Typical_sensitivity>
        <Sensitivity_specified_by_manufacturer type='DBL'>0</Sensitivity_specified_by_manufacturer>
        <Reference_frequency type='DBL'>0</Reference_frequency>
      </Sensor_cluster_1>
      <Transfer_function dim='[1]' type='Cluster'>
        <Transfer_function mems='2'>
          <Frequency_Hz_-----------Sensitivity_pC_per_m_s2_ dim='[40,2]' type='DBL'>
            <Numeric_Control>5</Numeric_Control>
            <Numeric_Control>0.002009</Numeric_Control>
            <Numeric_Control>6.3</Numeric_Control>
            <Numeric_Control>0.00201</Numeric_Control>
            <Numeric_Control>8</Numeric_Control>
            <Numeric_Control>0.002011</Numeric_Control>
            <Numeric_Control>10</Numeric_Control>
            <Numeric_Control>0.002011</Numeric_Control>
            <Numeric_Control>12.5</Numeric_Control>
            <Numeric_Control>0.002011</Numeric_Control>
            <Numeric_Control>16</Numeric_Control>
            <Numeric_Control>0.002012</Numeric_Control>
            <Numeric_Control>20</Numeric_Control>
            <Numeric_Control>0.002012</Numeric_Control>
            <Numeric_Control>25</Numeric_Control>
            <Numeric_Control>0.002013</Numeric_Control>
            <Numeric_Control>31.5</Numeric_Control>
            <Numeric_Control>0.002013</Numeric_Control>
            <Numeric_Control>40</Numeric_Control>
            <Numeric_Control>0.002013</Numeric_Control>
            <Numeric_Control>50</Numeric_Control>
            <Numeric_Control>0.002013</Numeric_Control>
            <Numeric_Control>63</Numeric_Control>
            <Numeric_Control>0.002013</Numeric_Control>
            <Numeric_Control>80</Numeric_Control>
            <Numeric_Control>0.002014</Numeric_Control>
            <Numeric_Control>100</Numeric_Control>
            <Numeric_Control>0.002014</Numeric_Control>
            <Numeric_Control>125</Numeric_Control>
            <Numeric_Control>0.002014</Numeric_Control>
            <Numeric_Control>160</Numeric_Control>
            <Numeric_Control>0.002015</Numeric_Control>
            <Numeric_Control>200</Numeric_Control>
            <Numeric_Control>0.002015</Numeric_Control>
            <Numeric_Control>250</Numeric_Control>
            <Numeric_Control>0.002015</Numeric_Control>
            <Numeric_Control>315</Numeric_Control>
            <Numeric_Control>0.002016</Numeric_Control>
            <Numeric_Control>400</Numeric_Control>
            <Numeric_Control>0.002016</Numeric_Control>
            <Numeric_Control>500</Numeric_Control>
            <Numeric_Control>0.002017</Numeric_Control>
            <Numeric_Control>630</Numeric_Control>
            <Numeric_Control>0.002017</Numeric_Control>
            <Numeric_Control>800</Numeric_Control>
            <Numeric_Control>0.002018</Numeric_Control>
            <Numeric_Control>1000</Numeric_Control>
            <Numeric_Control>0.002019</Numeric_Control>
            <Numeric_Control>1250</Numeric_Control>
            <Numeric_Control>0.002021</Numeric_Control>
            <Numeric_Control>1600</Numeric_Control>
            <Numeric_Control>0.001968</Numeric_Control>
            <Numeric_Control>2000</Numeric_Control>
            <Numeric_Control>0.002018</Numeric_Control>
            <Numeric_Control>2500</Numeric_Control>
            <Numeric_Control>0.002024</Numeric_Control>
            <Numeric_Control>3150</Numeric_Control>
            <Numeric_Control>0.002031</Numeric_Control>
            <Numeric_Control>4000</Numeric_Control>
            <Numeric_Control>0.002043</Numeric_Control>
            <Numeric_Control>5000</Numeric_Control>
            <Numeric_Control>0.002054</Numeric_Control>
            <Numeric_Control>5500</Numeric_Control>
            <Numeric_Control>0.002059</Numeric_Control>
            <Numeric_Control>6300</Numeric_Control>
            <Numeric_Control>0.002074</Numeric_Control>
            <Numeric_Control>7000</Numeric_Control>
            <Numeric_Control>0.002092</Numeric_Control>
            <Numeric_Control>7500</Numeric_Control>
            <Numeric_Control>0.002102</Numeric_Control>
            <Numeric_Control>8000</Numeric_Control>
            <Numeric_Control>0.002114</Numeric_Control>
            <Numeric_Control>8500</Numeric_Control>
            <Numeric_Control>0.002132</Numeric_Control>
            <Numeric_Control>9000</Numeric_Control>
            <Numeric_Control>0.002146</Numeric_Control>
            <Numeric_Control>9500</Numeric_Control>
            <Numeric_Control>0.002162</Numeric_Control>
            <Numeric_Control>10000</Numeric_Control>
            <Numeric_Control>0.00218</Numeric_Control>
          </Frequency_Hz_-----------Sensitivity_pC_per_m_s2_>
          <Mass_gram_ type='DBL'>30</Mass_gram_>
        </Transfer_function>
      </Transfer_function>
      <Transfer_function_phase mems='1'>
        <Frequency_Hz_------------Phase_degree_ dim='[40,3]' type='DBL'>
          <Numeric_Control>5</Numeric_Control>
          <Numeric_Control>2.67</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>6.3</Numeric_Control>
          <Numeric_Control>2.12</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>8</Numeric_Control>
          <Numeric_Control>1.68</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>10</Numeric_Control>
          <Numeric_Control>1.35</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>12.5</Numeric_Control>
          <Numeric_Control>1.09</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>16</Numeric_Control>
          <Numeric_Control>0.87</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>20</Numeric_Control>
          <Numeric_Control>0.75</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>25</Numeric_Control>
          <Numeric_Control>0.58</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>31.5</Numeric_Control>
          <Numeric_Control>0.46</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>40</Numeric_Control>
          <Numeric_Control>0.37</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>50</Numeric_Control>
          <Numeric_Control>0.31</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>63</Numeric_Control>
          <Numeric_Control>0.26</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>80</Numeric_Control>
          <Numeric_Control>0.22</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>100</Numeric_Control>
          <Numeric_Control>0.18</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>125</Numeric_Control>
          <Numeric_Control>0.17</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>160</Numeric_Control>
          <Numeric_Control>0.16</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>200</Numeric_Control>
          <Numeric_Control>0.11</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>250</Numeric_Control>
          <Numeric_Control>0.09</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>315</Numeric_Control>
          <Numeric_Control>0.08</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>400</Numeric_Control>
          <Numeric_Control>0.07</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>500</Numeric_Control>
          <Numeric_Control>0.06</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>630</Numeric_Control>
          <Numeric_Control>0.04</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>800</Numeric_Control>
          <Numeric_Control>0.03</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>1000</Numeric_Control>
          <Numeric_Control>0.02</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>1250</Numeric_Control>
          <Numeric_Control>0</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>1600</Numeric_Control>
          <Numeric_Control>-2.39</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>2000</Numeric_Control>
          <Numeric_Control>-0.03</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>2500</Numeric_Control>
          <Numeric_Control>-0.05</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>3150</Numeric_Control>
          <Numeric_Control>-0.09</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>4000</Numeric_Control>
          <Numeric_Control>-0.14</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>5000</Numeric_Control>
          <Numeric_Control>-0.37</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>5500</Numeric_Control>
          <Numeric_Control>-0.3</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>6300</Numeric_Control>
          <Numeric_Control>-0.31</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>7000</Numeric_Control>
          <Numeric_Control>-0.28</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>7500</Numeric_Control>
          <Numeric_Control>-0.33</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>8000</Numeric_Control>
          <Numeric_Control>-0.31</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>8500</Numeric_Control>
          <Numeric_Control>-0.42</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>9000</Numeric_Control>
          <Numeric_Control>-0.4</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>9500</Numeric_Control>
          <Numeric_Control>-0.42</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
          <Numeric_Control>10000</Numeric_Control>
          <Numeric_Control>-0.42</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
        </Frequency_Hz_------------Phase_degree_>
      </Transfer_function_phase>
      <Maximum_payload mems='1'>
        <From_Frequency_Hz_------To_Frequency_Hz_------max.Payload_gram_ dim='[1,3]' type='DBL'>
          <Numeric_Control>5</Numeric_Control>
          <Numeric_Control>10000</Numeric_Control>
          <Numeric_Control>100</Numeric_Control>
        </From_Frequency_Hz_------To_Frequency_Hz_------max.Payload_gram_>
      </Maximum_payload>
      <Sensor_tolerance mems='3'>
        <lower_frequency_Hz_------higher_Frequency_Hz_--max.deviation_percent_--min.deviation_percent_ dim='[4,3]' type='DBL'>
          <Numeric_Control>5</Numeric_Control>
          <Numeric_Control>20</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>20</Numeric_Control>
          <Numeric_Control>1000</Numeric_Control>
          <Numeric_Control>0.3</Numeric_Control>
          <Numeric_Control>1000</Numeric_Control>
          <Numeric_Control>5000</Numeric_Control>
          <Numeric_Control>0.5</Numeric_Control>
          <Numeric_Control>5000</Numeric_Control>
          <Numeric_Control>10000</Numeric_Control>
          <Numeric_Control>1</Numeric_Control>
        </lower_frequency_Hz_------higher_Frequency_Hz_--max.deviation_percent_--min.deviation_percent_>
        <Maximum_in_percent type='DBL'>0.3</Maximum_in_percent>
        <Minimum_in_percent type='DBL'>0</Minimum_in_percent>
      </Sensor_tolerance>
      <Sensor_cluster_2 mems='26'>
        <Weight_gram_ type='DBL'>30</Weight_gram_>
        <Transverse_sensitivity_in_percent_ type='DBL'>0</Transverse_sensitivity_in_percent_>
        <Capacitance type='DBL'>0</Capacitance>
        <Offset_voltage type='DBL'>0</Offset_voltage>
        <Warm-up_time_in_seconds_ type='DBL'>0</Warm-up_time_in_seconds_>
        <Electrically_insulated type='String'></Electrically_insulated>
        <TEDS type='String'></TEDS>
        <ID_module type='String'></ID_module>
        <IDmodule_ID type='String'></IDmodule_ID>
        <Comment type='String'></Comment>
        <Date_of_purchase type='DBL'>0</Date_of_purchase>
        <History type='String'></History>
        <Is_this_a_permanently_monitored_measurement_device type='String'></Is_this_a_permanently_monitored_measurement_device>
        <Person_responsible type='String'></Person_responsible>
        <Location type='String'></Location>
        <Resonance_frequency type='DBL'>0</Resonance_frequency>
        <Date_of_change type='DBL'>0</Date_of_change>
        <Changed_by type='String'></Changed_by>
        <Date_of_last_calibration type='DBL'>3.629146E+9</Date_of_last_calibration>
        <Date_of_next_calibration type='DBL'>3.660682E+9</Date_of_next_calibration>
        <Calibration_interval_in_days type='DBL'>360</Calibration_interval_in_days>
        <Calibration_was_performed_by type='String'>Spektra</Calibration_was_performed_by>
        <Number_of_calibration_certificate type='String'>DK15183-01-00</Number_of_calibration_certificate>
        <Minimum_frequency type='DBL'>5</Minimum_frequency>
        <Maximum_frequency type='DBL'>10000</Maximum_frequency>
        <Maximum_frequency_for_sweep type='DBL'>10</Maximum_frequency_for_sweep>
      </Sensor_cluster_2>
    </Sensor>
  </Array>
</GXML_Root>